Dec 31, Operating and service manual for the HPA Synthesizer/Function Generator. Complete with schematics. Who I HEWLETT mllnM PACKARD OPERATING AND SERVICE MANUAL MODEL A SYNTHESIZER/FUNCTION GENERATOR Serial Numbers: AH. HP / Hewlett Packard / Agilent Used A Synthesizer/Function Generator, Refurbished – Used Test Equipment at Test Equipment Depot.
|Published (Last):||15 August 2014|
|PDF File Size:||13.11 Mb|
|ePub File Size:||17.90 Mb|
|Price:||Free* [*Free Regsitration Required]|
If your oscilloscope does not have a ohm input, use a ohm load -hp- Model C ohm Feedthru Termination at the input. Add – 6 dB to this number and enter on the performance test card. Change the trigger on the oscilloscope to positive and note the negative ramp jitter in microseconds.
Set the digital voltmeter delay to. In order to use the Oven Option The tilt stand raises the front of the instrument for easier viewing of the control panel. A list of these offices is provided at the end of this manual.
No adjustments can be made.
Full text of “HP A Synthesizer/Function Generator Operating and Service Manual”
Option is a temperature stabilized 10 MHz oscillator which provides improved frequency stability see specifications in Table Test instrument parameters Harmonic distortion relative to fundamental at full output Amplitude modulation Square wave Ramp retrace Frequency Sine wave Square wave Triangle Positive slope ramp Negative slope ramp Phase offset Amplitude Amplitude accuracy with no attenuation attenuator range 1 into 50 load no dc offset Flatness with no attenuation As the modified digit passes 9 incrementing or 0 decrementing the digit to its left will increment or decrement.
Connect the equipment as shown in Figure 4- manyal.
No circuit or comp, layout changes. Valid EOS characters are: To print the manual completely, please, download it. Wrap the instrument in heavy paper or plastic. Change the dc offset to NOTE Only one signal output is active at one time. The following backdating information is organized by service group with applicable information placed together for easy reference.
Using the Modify manusl, adjust the phase until the counter reads approximately nanoseconds. This is a basic program statement which accomplishes the following: Change A amplitude to 1 V p-p, and change oscilloscope vertical to.
The power cable must either be plugged into an approved three-contact electrical outlet or used with a three-contact to two-contact adapter with the grounding wire green firmly connected to an electrical ground safety ground at the power outlet.
Frequency Accuracy 4 Set oscilloscope controls to measure ramp retrace time from the 90 to 10 percent points.
3325A Synthesizer/Function Generator
The following are valid programming strings data messages for the A: This printout results when string variables are used. The supplement should be manusl with the instrument for use by the operator.
Press the key in the lower right corner of the A front panel to select the High Voltage output. Write the program in flowchart or algorithm form. After entering a phase shift, the new phase may np assigned the zero phase position, and subsequent changes in phase referenced to that point.
HP A CALIBRATION MANUAL Pdf Download.
Low at start of sweep, High at stop. The instructions outlined in paragraphs 6 and 7 are preparatory to the calibration process. Page 22 TB Figure 5. In both cases the controller places the bus in the command mode to execute the message. To obtain replacement parts, address order or inquiry to your local Hewlett-Packard Field Office.
The error decreases as sweep lime is increased. This output is identical to the output at the front panel sync connector. These instruments do not have R If the calculator is equipped with a String Variable ROM, the interrogate program may be changed to the following. Valid data for SM is: Front and Rear Panel Controls .
Verify Vpp to be between and volts.